Hebei Huachuang M&C Technology Co.,LTd You are welcome！ Please contact customer service directly dial the right number！
Focus On Level Measurement
Radar level meter installed on the horizontal tank what is easy to be ignored
When it comes to the installation of radar level meter, many users do not know, because the installation is prone to problems. Some problems are invisible and usually not found at all.
Radar level meter adopts advanced microprocessing technology, which can be applied to a variety of complex industrial production, and its process connection and type are very special, which can be applied to various occasions with high measurement accuracy. Before using the radar level meter, you need to install the radar level meter. It is important to install the radar level meter, but if you don't pay attention, it is easy to overlook some details, resulting in inaccurate radar level meter measurement results. The reasons for inaccuracy are various, man-made and the quality itself. Here we focus on the installation of radar level meter on the horizontal tank is easy to be ignored.
Radar level meter in the installation direction, the level of the radar level meter can not enter the measurement blind area, once exceeded, will have a negative impact on the radar level meter. In addition, the distance between the radar level meter and the horizontal tank should be maintained to avoid damage between the radar level meter and the horizontal tank. In addition, when installing the radar level meter, its direction can be perpendicular to the surface of the measured medium.
The tank wall requires the radar level meter to be as far away from the outlet and inlet as possible The radar level meter should not touch the horizontal tank wall or the bottom of the tank during the whole measuring range, which will affect the measuring accuracy of the radar level meter. In addition, there should not be any obstacle within the measuring range, which will lead to inaccurate measurement.
Comply with explosion-proof requirements. When installing the radar level meter, you should ensure that the direction of the radar level meter is perpendicular to the surface of the measured medium as far as possible. If we use radar level meter in explosion-proof area, we need to install it in accordance with relevant regulations and meet the explosion-proof requirements.